NSF Certified 6 Ft Class II Type A2 Biosafety Cabinet
Our biosafety cabinet is a critical need product intended to serve special needs markets. Strictly MSRP pricing only applies.
A biosafety cabinet, also called a biological safety cabinet or microbiological safety cabinetis an enclosed, ventilated laboratory workspace for safely working with materials contaminated with (or potentially contaminated with) pathogens requiring a defined biosafety level.
The primary purpose of a BSC is to serve as a means to protect the laboratory worker and the surrounding environment from pathogens. All exhaust air is HEPA-filtered as it exits the biosafety cabinet, removing harmful bacteria and viruses. This is in contrast to a laminar flow clean bench, which blows unfiltered exhaust air towards the user and is not safe for work with pathogenic agents. Neither are most BSCs safe for use as fume hoods. Likewise, a fume hood fails to provide the environmental protection that HEPA filtration in a BSC would provide. However, most classes of BSCs have a secondary purpose to maintain the sterility of materials inside (the "product").
Class I cabinets provide personnel and environmental protection but no product protection. In fact, the inward flow of air can contribute to contamination of samples. Inward airflow is maintained at a minimum velocity of 75 ft/min(0.38 m/s). These BSCs are commonly used to enclose specific equipment (e.g. centrifuges) or procedures (e.g. aerating cultures) that potentially generate aerosols. BSCs of this class are either ducted (connected to the building exhaust system) or unducted (recirculating filtered exhaust back into the laboratory.
Class II cabinets provide both kinds of protection (of the samples and of the environment) since makeup air is also HEPA-filtered. There are five types: Type A1 (formerly A), Type A2 (formerly A/B3), Type B1, Type B2 and Type C1. Each type's requirements are defined by NSF International Standard 49, which in 2002 reclassified A/B3 cabinets (classified under the latter type if connected to an exhaust duct) as Type A2, and added the Type C1 in the 2016 standard. About 90% of all biosafety cabinets installed are Type A2 cabinets.
Ai Class II Type A2 biosafety cabinet features
- NSF certified
- Great for COVID-19 specimen handling
- Motorized front glass viewing window with optional foot switch
- Dual HEPA filters with life indicator
- UV light life indicator
- Waterproof in-chamber power sockets
- Large LCD display
- Easy-to-clean stainless steel chamber
- Special treated unit housing with anti-mold and antiseptic coating
- Auto start and stop function (e.g. UV lights)
- Automatic flow speed adjustment
- Pneumatic front panel supports for easy maintenance
IMPORTANT GUIDELINES USING A BSC
- No flammable and explosive substances (Building up of flammable material may cause an internal fire or explosion)
- No volatile chemicals (May cause damage to HEPA filters to the point where it allows particles to leak through)
- No toxic chemicals or materials (May reach room occupants)
- No open flame (Will disrupts air flow patterns)
- Minimal use (10mL or less) of decontaminant, such as 70% isopropyl or ethyl alcohol is allowed.
- We recommend using the following for disinfecting the surfaces of a BSC, depending on the agent to be controlled: Sodium hypochlorite (bleach), Ilodophors (Wescodyne), Phenolics (Hilphene, Vespene) and Quaternary ammonium compounds (Conflikt, EndbacII)
- If gaseous, vaporized or aerosolized chemicals must be generated in a BSC, you should assure yourself that the quantity being generated will NOT cause a flammable or toxic risk.